Int. J. Metrol. Qual. Eng.
Volume 9, 2018
|Number of page(s)||6|
|Published online||29 October 2018|
Temperature effect in the calibration of capacitive humidity sensors
Instituto Nacional de Metrologia de Colombia,
* Corresponding author: firstname.lastname@example.org
Accepted: 26 September 2018
Capacitive hygrometers are widely used in industrial and environmental measurements. One of the limitations of these sensors is their temperature dependence. While in temperatures beyond 50 °C the effect has been proven to be significant, for standard conditions in calibration laboratories the magnitude of this dependence has not been rigorously studied. This paper presents the analysis and results of a study of the temperature influence in the calibration of capacitive hygrometers designed for monitoring environmental conditions. Ninety measurements were taken on 15 models of capacitive hygrometers using a two-pressure humidity generator between 30%rh and 80%rh and temperatures from 20 to 30 °C. A three-way analysis of variance (ANOVA) was used to evaluate the data. The results show that the calibration correction is independent of the temperature for the conditions considered; however, there is a correlation between the temperature and the devices, indicating that the gradients in temperature affect each technology of fabrication in a different way.
Key words: temperature effect / capacitive hygrometers / ANOVA / relative humidity / calibration
© S.A. Carvajal and C.A. Sánchez, published by EDP Sciences, 2018
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.