Int. J. Metrol. Qual. Eng.
Volume 9, 2018
|Number of page(s)||6|
|Published online||18 December 2018|
Extending dew-point temperature scale up to +50 °C
National Institute of Standard, Giza, Egypt
* Corresponding author: email@example.com
Accepted: 20 November 2018
Extension of dew-point temperature scale has been performed using a two-temperature (2-T), constant pressure humidity generator that is developed for the first time by the National Institute of Standard (NIS) in order to extend the calibration capabilities to the high dew-point temperature range at NIS. It relies on the saturation of a stream of gas flowing over a water surface maintained at constant, well-known, temperature. In this paper, primary realization of dew-point temperature scale in a dew-point temperature range up to +50 °C was performed to extend calibration capabilities and to improve the uncertainties of the dew-point temperature scale realization. Several experiments were carried out in order to characterize the generator. Characterization comprises studies of the saturator efficiency, temperature stability and a comparison with a calibrated chilled-mirror hygrometer. The results of the efficiency tests showed good performance of the generator. For uncertainty of measurements, a thorough analysis was also described representing estimations of contributions for all the sources that possibly affect measurements.
Key words: dew-point / hygrometer / two-temperature generator / uncertainty
© D.M.A. El-Gelil and M.G. Ahmed, published by EDP Sciences, 2018
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.