Mass of water as a function of the burner power.
|Burner power about the Hs in (kW)||Internal diameter of container in (mm)||Body of water M1 in (kg)|
|Between 1.16 and 1.64||220||3.7|
|Between 1.64 and 1.98||240||4.8|
|Between 1.99 and 2.36||2601)||6.1|
|Between 2.37 and 4.20||2601),2)||6.1|
1)If the container with a diameter of 260 mm cannot be used, the test must be conducted under the normal conditions of use of the appliance, using the 240 mm container with corresponding M1 water mass, and the burner power must be adjusted to 1.98 kW (on Hs).
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.