Acceptance criteria established in MSA manual.
|Bias and linearity||In general, significantly equal to zero and not exceed the maximum permissible error established by the gage calibration procedure.|
|%GRR||<10% tolerance (or process variation): Generally considered to be an acceptable measurement system.|
|10–30% tolerance (or process variation): May be acceptable for some applications, but should be approved by the customer.|
|Number of distinct categories (ndc)||Should be greater than or equal to 5.|
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.