Frequencies of lasers under calibration measured by CEM2 and frequency comb printed as a difference from nominal frequency values. All values in MHz.
|Type of lasers||Nominal frequency||Deviation from nominal frequency (with CEM2)||Deviation from nominal frequency (with comb)|
|HP5517D||473 612 254||22.76 ± 1.2||22.79 ± 1.0|
|HP5519A||473 612 254||19.40 ± 9.5||19.10 ± 0.9|
|TESA-RSD/SR||473 612 750||9.99 ± 0.5||9.98 ± 0.3|
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.