Int. J. Metrol. Qual. Eng.
Volume 7, Number 1, 2016
|Number of page(s)||6|
|Published online||19 April 2016|
New adiabatic calorimeter for realization the triple point of water in metallic-sealed cell at NIS-Egypt
National Institute for Standards, Tersa St., El Haram, Giza 12211, Egypt
⋆ Correspondence: email@example.com
Received: 30 November 2015
Accepted: 9 March 2016
The thermal metrology laboratory, at the National Institute for Standards (NIS-Egypt), has been working during the last three years on the development of metallic-sealed cells for the realization of the triple point of water (TPW), 273.16 K [H. Preston-Thomas, Metrologia 27, 3 (1990)]. A metallic-sealed cell has been already developed and characterized in the past [A. El Matarawy, M.G. Ahmed, Int. J. Metrol. Qual. Eng. 5, 401 (2014), M.G. Ahmed, A. El Matarawy, H.M. Abo Dorra, J. Phys. Sci. Appl. 1 (2015)]. The classical large triple point of water cell and metallic-sealed cell showed a discrepancy of 0.7 mK [A. El Matarawy, M.G. Ahmed, Int. J. Metrol. Qual. Eng. 5, 401 (2014)]. The main target of this work is to get the lowest possible uncertainty in calibrating Capsule Standard Platinum Resistance Thermometers (CSPRTs) under adiabatic conditions. The full system is composed of a metallic-sealed cell and an improved adiabatic calorimeter to increase as much as possible the thermal resistance between the cell and the inner adiabatic shield of the calorimeter. It gives also a reliable calculation of the heat capacity of the cell and the comparison copper block which accommodates several CSPRTs. With this new system the reproducibility of metallic-sealed cell was found to be 0.7 mK as described below. reproducibility of metallic-sealed cells is found to be less than 0.7 mK.
Key words: Adiabatic calorimeter / triple point of water (TPW) / metallic cell / CSPRT / uncertainty
© EDP Sciences 2016
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.